An off-the-shelf 30-cm TDR probe from Campbell Scientific was tested under laboratory conditions using a range of controlled air:water ratios, both in calm and turbulent conditions.
Because rivers are highly turbulent, two tests were carried out to determine if turbulence affects the performance of the TDR probe. In the first, the probe was sequentially immersed in calm and turbulent air. In the second, it was sequentially immersed in calm and turbulent water.
To obtain a regression between signal frequency and air content, the TDR probe was incrementally immersed into a calm water reservoir. This was done for both 30-cm and 10-cm wave guide configurations.
A turbulent air:water mixture calibration was created for the 30-cm probe by incrementally immersing the rate of air injected through a porous plate into the bottom of a cylindrical acrylic bubble chamber for water contents of 65 - 100 %.
- Air Content Measurements -
Lab Methods - Single
Phase Test Results - Air-Water Mixture
Calibrations - Turbulent Air-Water
Mixture Time Series
Field Methods - Field Site - Field Results
Please send comments about this WWW site to: firstname.lastname@example.org
Copyright © 1999. Gregory B. Pasternack and Brett L. Vallé. All Rights Reserved.